The handheld manometer, Tracoe Cuff Pressure Monitor Sensitive (CPMS), is used for accurate pressure control and for the filling of high-volume low-pressure cuffs of tracheostomy tubes and endotracheal tubes. The cpms can also be used to release air from the cuff. It is characterized by its clear scale, easy and quick measurement and a green marked field showing the normal pressure range. A separately available connecting tube with Luer lock connection can be attached additionally.
- TRACOE cuff pressure monitor (CPM) is used to fill and monitor the pressure of high-volume-low-pressure cuffs of tracheostomy and endotracheal tubes.
Units outstanding features and benefits: easy user precise, reproducible pressure indication extra-large, easy-to-read scale from 0 – 100cmH2Orn widely spaced scale markings with color-coded pressure range stable metal hook for attachment.
- The subglottic secretion suction to reduce the VAP rate of patients with an expected ventilation period of more thanrn72 hours was classified as category IA of the recommendations by the German Commission for Hospital Hygiene and Infectious Disease Prevention (guidelines for hospital hygiene and infection prevention issued by the Robert Koch Institute, KRINKO).
- Thernrecommendation specifies that the cuff pressure level should be set between 20 and 30 cmH2O and checked at regular intervals.
- The recommendation is part of category IB. When these two actions are taken in combination, they can help to reduce the incidence of pneumonia-TRACOE technic Connection Tube REF 702rn 1 m length with check valve